Portable Hardness Ultrasonic Test Machine Instrument
|FOB Price:||Get Latest Price|
|Min. Order:||1 Set|
|Production Capacity:||20 Sets/Month|
|Payment Terms:||L/C, T/T, Western Union|
- Model NO.: SVR-110
- Measuring Range: HV
- Standard: ISO, ASTM
- Mobility: Portable
- Rated Voltage: 220V/50Hz; 110V/60Hz
- Customized: Non-Customized
- Measuring Rang: Hv:50 ~ 999; HRC:20 ~ 70; Hb:85 ~ 650
- Hardness Scale: Hv, Hb, HRC, Hra, Hrb
- Instrument Dimensions: 160X80X31mm
- Data Storage: to Save 2000-Group of Measuring Data
- Accuracy: Hv:3%Hv; HRC:1.5HRC; Hb:3%Hb
- Transport Package: Carton Box
- Origin: Dongguan, China
- Type: Vickers Hardness Tester
- Application: Metal
- Warranty: 1 Year
- Calibrating Method: Standard Hardness Test Blocks
- Support Language: English
- Loading Force: Motorized Probe 2kgf
- Indenter: 136°vickers Diamond Indenter
- Net Weight: Approximately 500g ( Without Probe )
- Operating Voltage: DC 6V
- Measuring Direction: Support 360deg
- Trademark: SINOWON
- Specification: CE
- HS Code: 9024102000
High Measurement Accuracy - ±3% HV, ±1.5HR, ±3%HB;
Small Indentation - Only high-power microscope can observe the indentation;
Efficient Measurement - Result In 2 Seconds;
Large LCD - Directly display measurement result, times count, Maximum, Minimal, Average and deviation;
Easy Operation - Operate well after Short training;
High performance - 2-Year Warranty for Main Unit (Excludes Probe);
Mass Storage - Save 1000 groups measurement data;
Simple Calibration - Save 20 groups calibration data for invoking, improve calibration efficiency.
SU-100 Application Range:
At present, there are kinds of methods for hardness measurement, commonly used like Brinell, Rockwell, Vickers, Leeb, etc. while the disadvantages are obvious for the above hardness measurement.
Rockwell and Brinell with heavy loading force and big indentation, lead to serious destruction on sample surface.
Vickers apply optical measurement, but only professional technicians can operate smoothly, impossible to measure hardness of heavy work piece, installed machinery and permanently assembled parts.
Leeb hardness tester apply rebound and indirect method to measure hardness, easily lead to big deviation when convert to Birnell, Rockwell and Vickers scales.
Ultrasonic hardness tester SU100 apply ultrasonic contact impedance method to do comparative hardness measurement for testing pieces, with advantages of high accuracy, efficiency, portable and easy operation.
Ultrasonic hardness tester SU100 is widely used to measure hardness of small forgings, cast material, weld inspection, heat affected zone, Ion-nitrided stamping dies and molds, forms, presses, thin walled parts, bearings, tooth flanks, etc.
testing method developed by Dr. Claus Kleesattel in 1961 based on the measurement of
the frequency shift of a resonating rod caused by the essentially elastic nature of the finite
area of contact between the indenter and the test piece during the penetration, the execute
standard is ASTM A1038.
When selecting a hardness tester for your application, it is important for you to consider the following:
A, Choose the correct test method based on the application.
Plan to use the highest test force and largest indenter possible. Consider the effects of the shape and dimensions of your test sample.
B, Verify the test results meet your requirements for accuracy and repeatability. There are significant differences between levels of performance within each classification of tester. A difficult job on one tester could be very simple and fast on another.
Answer these key questions:
1, What kind of hardness scale to be used?
2, What is the material to be tested, and is this material suitable to the type of test method you are considering?
3, How large is the part, component or specimen to be tested?
4, Is the test point difficult to reach?
5, What is the quantity of testing that will be done?
6, How accurate does your test result need to be?
7, What is your budget?
8, What is the required return on investment and do you have ways to measure reductions in costs- yields, throughput, operator efficiency?
9, What testing problems have you experienced in your current method?
10, How knowledgeable are the users of the tester?
|Product Name||Ultrasonic Hardness Tester|
|Loading Force||2Kgf(Optional: 1Kgf, 5Kgf, 10Kgf)|
|Measuring Rang||HV:50 ~15 99; HRC:20 ~ 70; HB:85 ~ 650|
|Measuring Accuracy||HV:±3%HV; HRC:±1.5HRC; HB:±3%HB|
|Indenter||136°Vickers Diamond Indenter|
|Measuring Direction||Support 360°|
|Data Storage||To save 2000-group of measuring data and 20-group of calibration data|
|Data display||Loading force, testing-times, Testing result, Average, Maximum\Minimum\deviation and conversion scale.|
|Hardness indication||LCD display|
|Operating Environment||Temperature:-10ºC~50ºC; Humidity: 30%~80%R.H|
|Operating Voltage||DC 6V|
|Net Weight||Approximately 500g(Without probe)|
|Standard Delivery||Code#||Standard Delivery||Code#|
|Instrument Main Body||882-121||USB Cable||882-851|
|2Kgf manual Probe HP-2k||882-321||Accessories Box||882-901|
|Probe Silicone Cap||882-711||Instrument Manual||SU-100|
|Optional Accessories:||Code#||Optional Accessories:||Code#|
|1Kgf manual Probe HP-1k||882-311||Plan Support Ring||882-511|
|5Kgf manual Probe HP-5k||882-331||Small Cylinder Support Ring||882-521|
|10Kgf manual Probe HP-10k||882-341||Big Cylinder Support Ring||882-531|
|Deep Hole Probe Cap||882-721||Handy Silicone Shell||882-861|
|Hardness Block HRC||882-611||Mini-Printer||882-831|
|Hardness Block HV||882-651||Testing Stand MU-100||882-301|
|Manual Probe Specifications:|
|Oscillating Rod Diameter||2.4mm||2.4mm||3mm||3mm|
|Roughness of measuring surface||Ra<3.2um||Ra<5um||Ra<10um||Ra<15um|
|Motorized Probe Specifications:|
|Oscillating Rod Diameter||2.4mm||2.4mm||3mm||3mm|
|Min weight of test material||0.3kg||0.3kg||0.3kg||0.3kg|
|Support Ring Specifications:|
|Support Ring Name||Plan Support Ring||Small Cylinder Support Ring||Big Cylinder Support Ring|
|Application||Plan Test Piece||Diameter 8-22mm Test Piece||Diameter 16-80mm Test Piece|