Color Screen Ultrasonic Leeb Portable Hardness Lab Equipment
|FOB Price:||Get Latest Price|
|Min. Order:||1 Set|
|Production Capacity:||20 Sets/Month|
|Payment Terms:||L/C, T/T, Western Union|
- Model NO.: SH-500
- Measuring Range: HLD
- Standard: ISO, ASTM
- Mobility: Portable
- Rated Voltage: 220V/50Hz; 110V/60Hz
- Customized: Non-Customized
- Data Storage: Max 2000 Groups Test Results
- Hardness Indication: 240X160 Graphic DOT Color LCD
- Shell Material: Metal
- Language: Chinese, English, German, Portuguese, Turkish;Czec
- Statistics: Average Value, Max, Min, etc
- Transport Package: Carton Box
- Origin: Dongguan, China
- Type: Leeb Hardness Tester
- Application: Metal
- Warranty: 1 Year
- Calibrating Method: Standard Hardness Test Blocks
- Support Language: English
- Measuring Direction: Support 360deg
- Data Output: RS232/Mini-Printer
- Hardness Scales: Hl, Hbw, Hrb, HRC, Hra
- Indicating Error: Hld +/-6; HRC +/-1; Hb+/-4
- Impact Device: D as Standard
- Trademark: SINOWON
- Specification: CE
- HS Code: 9024102000
Die cavity of molds
Bearings and other parts
Failure analysis of pressure vessel, steam generator and other equipment
Heavy work piece
The installed machinery and permanently assembled parts
Testing surface of a small hollow space
Material identification in the warehouse of metallic materials
Rapid testing in large range and multi-measuring areas for large-scale work piece
SH-500 Leeb hardness tester features:
1,Metal shell, nice appearance design
2, Available to save 2000 groups test results
3, Support hardness conversion among HRC, HRA, HRB, HV, HB, etc, can convert 2 different scales
4, 4" LCD to display hardness result, conversion results, Max, Min, Mean value.
5, Support blue tooth printer, with RS232 to connect computer for software running
6, Can test all kinds of metal which more than 5kg
7, Indicating error ±5HLD, conform standard ASTM A956.
8, Support 7 types of impact devices
When selecting a hardness tester for your application, it is important for you to consider the following:
A, Choose the correct test method based on the application.
Plan to use the highest test force and largest indenter possible. Consider the effects of the shape and dimensions of your test sample.
B, Verify the test results meet your requirements for accuracy and repeatability. There are significant differences between levels of performance within each classification of tester. A difficult job on one tester could be very simple and fast on another.
Answer these key questions:
1, What kind of hardness scale to be used?
2, What is the material to be tested, and is this material suitable to the type of test method you are considering?
3, How large is the part, component or specimen to be tested?
4, Is the test point difficult to reach?
5, What is the quantity of testing that will be done?
6, How accurate does your test result need to be?
7, What is your budget?
8, What is the required return on investment and do you have ways to measure reductions in costs- yields, throughput, operator efficiency?
9, What testing problems have you experienced in your current method?
10, How knowledgeable are the users of the tester?
Verify the test results meet your requirements for accuracy and repeatability. There are significant differences between levels of performance within each classification of tester. A difficult job on one tester could be very simple and fast on another.
So, although hardness testers within a test method and classification look alike, there are many features that can significantly affect productivity and accuracy. A good example of features affecting performance is demonstrated in bench Rockwell hardness testing systems.
All can handle moderately long parts using larger anvils or jack rests, however the portable type can usually test large parts more quickly and accurately, when compared to other bench testers, which require external support stands or fixtures. The SHR150 series, on the other hand, is much easier to use on small, awkward parts.
|Portable Leeb Hardness Tester|
|Measuring Range||(170-960)HLD,(17-68.5)HRC,(19-651)HB,(80-976)HV, (30-100)HS,(59-85)HRA(13 -100)HRB|
|Data Storage||maximum 2000 Groups test Results|
|LCD display||240x160 graphic dot color LCD|
|Hardness Scale||HL, HRC ,HRB, HRA, HV, HB, HS|
|Indicating Error||HLD:±5; HRC:±1; HB:±4|
|Language Options||CHINESE, English, German, Portuguese, Turkish|
|Statistics||Average value, min-max ,upper-lower limits|
|Impact Device||D (standard)|
|Optional impact Devices||DC/D+15/DL/G/C/E|
|Work piece radius (convex/concave)||Rmin=50mm(with support ring Rmin=10mm)|
|Min. Workpiece weight||2~5kg on stable support 0.05~2kg with compact coupling except C and G impact device|
|Workpiece min. thickness couple||5mm,expect with impact device G:10mm,C:1mm|
|Workpiece min. case hardened depth||0.08mm except C impact device 0.2mm and impact device 1.2mm|
|Operating Temperature||0 deg to 40 deg|
|Weight||500g (Excluding impact device and printer)|
|RS-232 Data cable||852-810|
|Instrument Main Body||852-110|
|Impact device type D||852-410|
|Software Data Pro||852-380|
|Optional Impact device type||DL, DC, D+15, C, E, G|
|Optional Support Ring||852-510|